There’s no accounting for weather- and magnetic storms are no exception. Even on an average day, background variation caused by atmospheric activity can obscure survey data and make it harder to interpret. On a bad day, noisy atmospheric conditions can render your data totally unusable. Unless you use a second, stationary magnetometer to compensate for both random and systematic background variation, that is.
The new, smaller, lighter, and even easier to set up Sentinel2 helps you cancel out errors and delivers more accuracy.
|> 10,000 nT/m
|18,000 to 120,000 nT
|1/5 seconds to 4/second. 1/sec recommended
|WiFi web interface (access point mode)
|External power operation
|Possible with the optional External Power cable
|Sentinel2 Operating Parameters
|Weight on land
|15.2 kg (33.5 lbs)
|Weight in water
|4.3 kg (9.5 lbs), Neutrally buoyant without tripod
|Height with tripod: 1.32m (52 in), Tube Diameter: 9.5cm (3.75 in)
|Maximum Surface Incline Angle for Deployment on tripod
|Magnetometer Cylinder Depth Rating
|-25C to +60C
|-60C to +70C
|Lithium-Ion, 8V, 92 Wh
|Battery Charge Time
|Approx. 8-10 hrs with the included 2A charger.